Research Article - Biomedical Research (2016) Volume 27, Issue 4

## Computational modeling of amperometric choline biosensor for neurochemical detection

**P Ravikumar**

^{1*}, S Arumugam^{2}and N Jeyashanthi

^{2}^{1}Department of Biomedical Engineering, Velalar college of Engineering and Technology, Erode -638012, India

^{2}Department of Computer Science Engineering, Nandha Engineering college, Erode -638052, India

- *Corresponding Author:
- P Ravikumar

Department of Biomedical Engineering, Velalar College of Engineering and Technology, India

**Accepted on **April 15, 2016

**Visit for more related articles at**Biomedical Research

### Abstract

The behavior of biosensors is based on the well-defined physical, chemical and biological reactions, which are expressed by nonlinear differential equations. The characteristics of diffusion reaction of choline biosensor are analyzed mathematically for wide range of input parameters such as initial concentration of choline and thickness of choline oxide enzyme layer. The mathematical simulations are done in MATLAB software using partial differential equation solver pdepe function. The analytical results are compared with numerically simulated results for various conditions to validate the model parameters. A numerical simulation result shows less than 2% deviation from analytical results for various substrate concentrations. For dimensionless diffusion distance x=0 and sigma=1, the dimensionless substrate and product concentrations are 0.72 and 0.18 when t=1 and changes exponentially with time.

## Keywords

Choline, Biosensor, Pdepe, Mathematical modeling, MATLAB.

## Introduction

Biosensor is a compact analytical device, used for detection of very minute amount of chemical elements present in the living systems [1]. Now a days biosensor plays a very important role in disease detection especially in neurochemical detection [2]. The concentrations of neurochemicals are very minute and difficult to detect due to poor sensitivity and selectivity of conventional detection methods [3,4]. Choline is an essential component for functioning of cell and precursor for acetylcholine. The neuro transmitter choline and acetylcholine are mainly involved in Alzheimer disease [5]. Biosensors are potentially useful for quick detection of chemical components. The development of these biosensing systems depends on a quantitative measurement of the enzyme activity before and after exposure to a target substrate [6].

In recent days amperometric based electrochemical biosensors are highly used for fast and accurate detection [7,8]. The operations of electrochemical sensors are very simple and never affect the host material [9,10]. Amperometric biosensors, produces the output current based on the sensing materials on the working electrode act as a catalyst and catalyze the redox reaction. During measurement, the electrode potential is kept constant while the current is monitored. Optimization of biosensor is the time consuming process, for that mathematical modeling is used to reduce the optimization time, cost and optimize the analytical characteristics of actual biosensor [11-13].

The nonlinear problems in enzyme diffusion for different types of biosensors are solved using various numerical methods [14] and analytical methods [15,16]. Mathematical Modeling of amperometric enzyme biosensor based on finite difference scheme was reported for glucose detection [13-19]. The mathematical simulation at transient and steady state conditions are carried out using the finite difference scheme technique. The amperometric biosensor response is numerically analyzed by changing the model parameters.

## Mathematical Modelling

The amperometric choline biosensor was constructed using three electrode system with platinum based choline working electrode, Ag/Agcl reference electrode and platinum counter electrode. The following enzyme reaction takes place in working electrode [5].

(1)

The enzyme reaction for the above equation can be written as

(2)

Where E is the choline oxidase enzyme, S is the choline
substrate, ES is a transitory complex assumed to be at a steady
concentration, and P is the hydrogen peroxide product. The k_{1},
k_{-1}, k_{2} are the reaction rate constants. The mass balance equation for the above reaction in the enzyme layer can be
written using Fick’s second law,

(3)

(4)

Where S and P represent concentrations of substrate and
product, D_{S} and D_{P} are the corresponding diffusion
coefficients. V_{max} is the maximum enzymatic rate when the
enzyme is fully saturated; K_{M} is the Michaelis Menten
constant. For enzyme reaction, the output current produced by
the sensor is

(5)

Where n is the number of electrons involved in charge transfer and F is the Faraday constant.

*Dimensionless form*

The following parameters are used to convert the above Equations 3 and 4 into normalized form.

The Equations 3 and 4 reduce to the following dimensionless form

(7)

(8)

σ^{2}, the Damkohler number (Da), compares the diffusion rate
through the enzyme layer with the enzyme reaction rate. If
Damkohler less than one, then the enzyme kinetics controls the
biosensor response.

The dimensionless initial and boundary conditions are

(9)

The closed form expression for the current density from Equation 5,

(10)

*Numerical simulation*

The partial differential Equations 7 and 8 for the corresponding initial and boundary conditions are solved numerically using pdepe function in MATLAB software [20]. The partial differential equations are solved using finite difference schemes. Numerical solutions are obtained for various parameters such as initial substrate concentrations, ration of reaction diffusion, and sigma for every value of time and space vector.

## Result and Discussion

*Model validation*

The analytical solutions are validated for specific set of values
[21]. In this paper the numerically simulated data are validated
through analytical solutions for output current density. The
following parameters are used for simulation F=96.486 C/
mmol, n=2, V_{max}=100 μmol, KM=3 mmol, d=100 μm,
DS=1x10^{-9} m^{2}/s [19,21].

The **Figure 1** shows the amperometric response of choline
biosensor for various substrate concentrations. The percentage
of deviation of analytical from simulated result is calculated
that is less than 2%. It conforms that, the modeled data are very
much similar to experimental data.

*Sensor response*

**Figure 2** shows the numerical simulation result of Equation 7. **Figure 2A** indicates the diffusion activity of the substrate with
respect to time for constant Damkohler number sigma=0.1 with
initial substrate concentration of 1. **Figure 2B** shows the
dimensionless substrate concentration for various levels
dimensionless choline level. S* is increased due to various
initial substrate content S0. It increases exponentially with
respect to time and distance.

The numerical simulation output of Equation 8 is shown in **Figure 3** shows the dimensionless product P* with respect to dimensionless enzyme thickness for various Damkohler
number and diffusion coefficients rate R. **Figure 3A** and **Figure
3B** shows that product concentration P* reduces when
diffusion coefficient ratio or time t increases.

## Conclusion

The one dimensional computational model of choline
biosensor was simulated for substrate conversion and various
parameters that influence the diffusion of substrate and
product. The Damkohler number (σ^{2}) mainly influences the
output current density. The numerical result of output current
was compared with analytical result, and there was no
significant difference between the numerical result and
analytical result. This study will help the researchers in
biosensor filed to study the behavior of biosensor before actual
experimentation.

## References

- VoDinh T. Biosensors and Biochips-BioMEMS: In: Biomedical Nanotechnology. Springer, Germany 2007; 1-20.
- Ferreira AAP, Uliana CV, de Souza Castilho M, Pesquero NC, Foguel MV, dos Santos GP, Fugivara CS, Benedetti AV, Yamanaka H. Amperometric Biosensor for Diagnosis of Disease: In: State of the Art in Biosensors-Environmental and Medical Applications. InTech Publisher, Croatia, 2013; 253-289.
- Baoyan Wu, Zhongmin Ou, Xiaojuan Ju, Shihua Hou. Carbon Nanotubes/Gold Nanoparticles Composite Film for the Construction of a Novel Amperometric Choline Biosensor. J Nanomaterials 2011; 46: 1-6.
- Perry M, Li Q, Kennedy RT. Review of recent advances in analytical techniques for the determination of neurotransmitters. Anal ChemActuator 2009; 653: 1-22.
- Mitchell KM. Acetylcholine and Choline Amperometric Enzyme Sensors Characterized in Vitro and in Vivo. Anal Chemistry 2004; 76: 1098-1106.
- Grieshaber D, MacKenzie R, Voros J, Reimhult E. Electrochemical Biosensors - Sensor Principles and Architectures. Sensors 2008; 8: 1400-1458.
- Hristov SM. Amperometric biosensors. Bulgarian Chem Communications 2008; 40: 204-210.
- Yogeswaran U, Shen-Ming Chen. A Review on the Electrochemical Sensors and Biosensors Composed of Nanowires as Sensing Material. Sensors 2008; 8: 290-313.
- Chen C, Xie Q, Yang D, Xiao H, Yingchun F, Yueming T, Shouzhuo Y. Recent advances in electrochemical glucose biosensors: a review. RSC Adv 2013; 3: 4473-4491.
- Mell LD, Maloy JT. A model for the amperometric enzyme electrode obtained through digital simulation and applied to the immobilized glucose oxidase system. Anal Chemistry 1975; 47: 299-307.
- Baronas R, Kulys J, Petrauskas K, Razumiene J. Modelling Carbon Nanotubes-Based Mediatorless Biosensor. Sensors 2012; 12: 9146-9160.
- Dzyadevych SV, Soldatkin AP, Korpan YI. Biosensors based on enzyme field-effect transistors for determination of some substrates and inhibitors. Anal BioanalChem 2003; 3: 496-506.
- Neykov A, Rangelova V. Mathematical Modeling of the Biosensor Systems. Biotechnology & Biotechnological Equipment 1998; 12: 100-109.
- Ganesan SP, Saravanakumar K, Rajendran L. Mathematical Modeling of Multienzyme Biosensor System. Int J Comput Mathematics 2014.
- Praveen T, Rajendran L. Theoretical Analysis of an Amperometric Biosensor Based on Parallel Substrates Conversion. ISRN Electrochemistry 2014.
- Anitha A, Anitha S, Rajendran L. A Kinetic Model for Amperometric Biosensor at Mixed Oxidase Enzyme. Int J Electrochem Sci 2014; 9: 990-1002.
- Gaidamauskaite E, Barona R. A Comparison of Finite Difference Schemes for Computational Modelling of Biosensors. Nonlinear Anal Modelling Control 2007; 12: 359-369.
- Fransaer J, Ammam M. Mathematical Modeling of the Amperometric Response to Glucose of Glucose Oxidase Films Deposited by AC-Electrophoresis. J Sensor Technol 2011; 1: 17-21.
- Yang WY, Cao W, Chung TS, Morris J. Applied Numerical Methods using MATLAB. John Wiley & Sons, Inc., 2005.
- Aseris V. Computational Modelling of Wire Biosensor with Competitive Substrates Conversion. Proceedings of the World Congress on Engineering 2014.
- Qiang WZ, Jia JZ, Wang K, Juan JX, Chen HY, Xia XH. Exploration of Two-Enzyme Coupled Catalysis System Using Scanning Electrochemical Microscopy. Anal Chem 2012; 84: 10586-10592.